The scale kit is diverse usable and has an excellent price-performance ratio.
The loadcells have a breaking load of 150 percent and are completely dust and waterproof according to protection class IP68. The feet are equipped with large-sized adjustment thread.
The accuracy of the loadcells is +/- 0,02 %.
If you order one of the AGRETO weighing indicators (HD1, XK3), then you get the components fully connected and calibrated. Your weighing technology is ready for immediate use.
• High-resolution shear beam load cells
• Capacity 1.000 kg
• 3 mV/V, 350 Ohm
• 120% secure overload, 150% breaking load
• +/- 0.02% accuracy
• Waterproof according to protection class IP68
• Operating temperature: -35 up to +65 °C
• Temperature compensated: -10 up to +40 °C
• Waterproof box
• Connecting terminals for weighing cells and weighing signal cable
• Potentiometer for corner adjustment of load cells
• Cable outlets with cable screw connections
• 4-pin cable 4 x 0.5 mm2
• 7.6 mm outer diameter
• Special coating
• Small bending radius
The AGRETO Scales kit with shear beam load cells can be used for any scale constructions. They are ideal for animal scales, for all kinds of platforms, for installation in machines such as fertilizer spreaders, seeders, wrappers, balers, forage mixers, trailers, etc. and also for mobile scale constructions.
To get a satisfactory result, the load cells has to be mounted as horizontally as possible. With the height-adjustable feet level differences can be compensated.
The load cells including the spacer plates are bolted with two high-strength screws on the platform structure normally from bottom to top. In the area of load cell mounting the scale construction must be resistant against bending and twisting. When utilizing the rated load, the material in the range of load cell mounting should have about half the thickness of the load cell itself. That is about 15 mm with a 1000 kg load cell. (see the pictures).
The load cells have two through holes on one side and a bore with a thread on the other side. The load cell must be mounted in that way, that the cell rests only in the region of the attachments to the structure and the entire load is transferred through the load cell. On the side with the two holes, the spacer plate can be used. On the side with the thread, a disc should be placed. To avoid tensions it is advisable to only tighten one cell and carry out the other fixtures floating. This is best done with a screw and a lock nut on the threaded side of the cell (-> see the pictures).
The distance between the loadcells is only limited with the length of the cables.
Please call us: +43 2846 620.